January 12, 2023 2:25:45 PM

Configuration

| Setup Configuration

| | Oscilloscope Version | | 5.3.5 Build 22
| | DPOJET Version | | 5.0.1 Build 8

| Measurement Configuration

| Index | Measurement | Source(s) | Others
| 1 | TIE1 | Ch2 | Edges => Signal Type: Auto, Clock Edge: Fall | Clock Recovery => Method: Constant Clock – Mean, Auto Calc: Every Acq, Nominal Data Rate: Off, Bit Rate: 2.5Gb/s, Known Data Pattern: Off, Pattern Filename: C:\Users\Public\Tektronix\Tekapplications\DPOJET\Patterns\PRBS127.txt | Filters => F1: Spec: No Filter, F2: Spec: No Filter | General => Measurement Range Limits: Off, Max: 1ns, Min: -1ns, Custom Source Name: fdiv(Ch2)
| 2 | Phase Noise1 | Ch2 | Edges => Active Edge: Fall, Noise Integration Limits: Upper Frequency: 1MHz, Lower Frequency: 0Hz | Filters => F1: Spec: No Filter, F2: Spec: No Filter | General => Measurement Range Limits: Off, Max: 1ms, Min: 0s, Custom Source Name: fdiv(Ch2)
| 3 | Period1 | Ch2 | Edges => Signal Type: Auto, Clock Edge: Rise | Filters => F1: Spec: No Filter, F2: Spec: No Filter | General => Measurement Range Limits: Off, Max: 1ms, Min: 0s, Custom Source Name: fdiv(Ch2)
| 4 | Freq1 | Ch2 | Edges => Signal Type: Auto, Clock Edge: Rise | Filters => F1: Spec: No Filter, F2: Spec: No Filter | General => Measurement Range Limits: Off, Max: 10GHz, Min: 1MHz, Custom Source Name: fdiv(Ch2)

| Source Reference Levels

| Source | Autoset Method | Rise High | Rise Mid | Rise Low | Hysteresis | Fall High | Fall Mid | Fall Low
| Ch1 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ch2 | Auto(Low-High(full wfm)) | 2.9252V | 1.61V | 294.8mV | 98.64mV | 2.9252V | 1.61V | 294.8mV
| Ch3 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ch4 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Math1 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Math2 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Math3 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Math4 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ref1 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ref2 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ref3 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V
| Ref4 | Auto | 1V | 0V | -1V | 30mV | 1V | 0V | -1V

| Miscellaneous Settings

| | Source Gating | Source Qualify | Stat Population Limit
| State | Off | Off, Active: High | Off, Limit By: Acquistions, Stop Condition: Each Measurement
| Source | -- | Ch4 | --
| Size | -- | -- | 1k

Measurement Results

| | Description | Mean | Std Dev | Max | Min | p-p | Population | Max-cc | Min-cc
| | TIE1, fdiv | 2.3750as | 1.0375ns | 4.7129ns | -4.3906ns | 9.1036ns | 224220 | 3.2593ns | -3.0982ns
| | Current Acquisition | -38.459as | 1.0615ns | 3.3568ns | -3.4825ns | 6.8392ns | 1515 | 2.1970ns | -2.1347ns
| | Phase Noise1, fdiv | 1.0604ns | 49.431ps | 1.1872ns | 931.53ps | 255.64ps | 148 | 0.0000s | 0.0000s
| | Current Acquisition | 1.0504ns | 0.0000s | 1.0504ns | 1.0504ns | 0.0000s | 1 | 0.0000s | 0.0000s
| | Period1, fdiv | 13.203us | 685.46ps | 13.205us | 13.199us | 5.9568ns | 223924 | 3.7898ns | -4.5611ns
| | Current Acquisition | 13.203us | 689.25ps | 13.205us | 13.200us | 4.6251ns | 1513 | 2.8625ns | -2.6413ns
| | Freq1, fdiv | 75.742kHz | 3.9324Hz | 75.761kHz | 75.727kHz | 34.175Hz | 223924 | 26.166Hz | -21.740Hz
| | Current Acquisition | 75.742kHz | 3.9542Hz | 75.756kHz | 75.729kHz | 26.534Hz | 1513 | 15.154Hz | -16.424Hz

Pass/Fail Summary

No pass/fail limits are currently selected.

Plot Images

| Measurement Plot(s)
|
|
|
| Oscilloscope Waveform

Reference Waveforms stored at

| | Waveform | Location
| | Ch2 | E:\DPOJET\2023.1\MeasReport1.56ghz_Ch2.wfm